Sample out of possible range WinDaq add-ons is the only UltimaSerial product being supported actively, all other software products are in FREEZE state, meaning no more modification nor active support. When Ultimaserial is first loaded, the channel allocation table is 1, 2, 3, Sorry, but the pattern of the square wave cannot be controlled. When you specify a sample rate, an attempt is made to report data at that sample rate. You cannot use PChannel , PGain property to change the channel and gain allocation table. The included samples are mostly from VB6.
|Date Added:||10 March 2016|
|File Size:||38.47 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
CommPort outside of range, which are from 0 to 64 for physical comm port, up to for USB-based virtual comm port. The specified integer must be between 0 and 32, NET, please also check dafaq one out:. You cannot use PChannelPGain property to change the channel and dxtaq allocation table. Lets you write the number of channels you want to be enabled or read the number of channels that are currently enabled.
Sorry, but the pattern of the square wave cannot be controlled.
Please use PChannelPGain property to change the channel and gain allocation table. When a specified PChannel passes the limit line, event LimitLine will be fired. Since the default gain of 0 is not a valid number, user must specify the range first before using the rate channel.
The availability of a digital inputs can be found in here. Failed to send key to DI In this case, you should use Timer to pace the acquisition Syntax: Like other ActiveX controls, UltimaSerial ActiveXs are bit components, you must select bit code option when using bit dztaq.
Our products –
Specify the COMM port serial port you want to use or find out which one you are currently using. MaskOutDigitalBits Boolean when set, the digital bits will be masked out from the data stream default: It serves as the entry point for the superkeys of site license Example: PreTriggerLength should be zero Applicable Devices: Invalid device name Channel Count changes to 1 When Ultimaserial is first loaded, the channel allocation table is 1, 2, 3, All It serves as the entry point for the superkeys of site license Example: Trigger Level too high, and will be changed to ei-158u Please refer to DI protocol or DI protocol for more detail.
Original channel count is unmodified Click here to find out how to extract the data from the variant. In this case, it will be much easier to use GetDataEx to place the data directly to a pre-defined integer array. Failed to create monitoring process, please poll the data adtaq VB Timer function SetLimitLine int type, int threshold specifies the type of monitoring and threshold to be used in LimitLine operation.
We ei-158u suggest DataqSDK.
[ubuntu] Trying to get a signal from an AD converter. [Archive] – Ubuntu Forums
DI is the default. This channel does not support alarm setting This method must be invoked before Start.
This returns the serial number of the device after you Start the device successfully. When this is not possible, the data will be reported at the closest achievable rate to the sample rate you requested.
When you specify the channel count in UltimaSerial, like: Invalid key to DI! This method must be invoked after Start Applicable Devices: